- Email: email@example.com
- Registered on: 11/18/2011
- Last connection: 11/11/2020
- llrf-hwr-8cav-fe-software (Manager, Developer, 08/10/2020)
- ANL Up/Down Converter Builds 2020/2021 (Manager, Developer, 08/25/2020)
- llrf_hwr_fe_software_acnet (Manager, Developer, 07/09/2020)
- VMEdsp (Manager, 10/19/2020)
- Main Injector and MRF (Manager, 03/11/2020)
- LLFrontends (Manager, 10/19/2020)
- 3.9 GHz Cryomodule testing (Manager, 02/01/2020)
- llrf-ssr1-fe-software-acnet (Manager, Developer, 08/14/2020)
- BOE and RLS (Manager, Developer, 04/10/2020)
- llrf-hwr-2cavity-labview (Manager, Developer, 08/28/2020)
- llrf-SSR1-fe-software (Manager, 10/16/2019)
- llrf-HWR-fe-software (Manager, Developer, 08/15/2019)
- LELPP (Manager, 11/18/2011)
- CMTS Q0 Extract (Manager, 07/26/2018)
- MEBT Chopper (Manager, 07/18/2018)
- PIP II IT SSR1 (Manager, Developer, 10/16/2019)
- llrf-SSR1-2Cavity-firmware (Manager, 10/16/2019)
- LBL LLRF System (Manager, Developer, 09/17/2020)
- PIP-II IT HWR (Manager, Developer, 08/15/2019)
- llrf-hwr-2cavity-firmware (Manager, 10/16/2019)
- 01:56 PM PIP II IT SSR1 Bug #25181 (New): Feedback gain ranges
- P gain is limited to 1000 and I gain is limited to 6E6. I think we could go higher although FB is working pretty well.
- 01:34 PM PIP II IT SSR1 Bug #25180 (New): Beam Loading Compensation timing values are off by a factor of 5
- BLC is five times longer than is indicated.
- 12:08 PM PIP II IT SSR1 Bug #25179 (New): SSR1 Freq Track Error
- If the Freq Er selector is set there is no readback in waveform or readback. In det Er (light off) the waveform is p...
- 11:48 AM PIP II IT SSR1 Bug #25178 (New): Labview FFT
- The FFT is only single sided and we need the full analytic signal shown. Also, we need to see all RF signals.
- 10:12 AM llrf-HWR-fe-software Bug #25028 (Closed): Netmask is not correct
- From Greg Brown -
To start off with.
From my studies I suspect that the following nodes are using a 16 bit...
- 05:18 PM llrf-hwr-2cavity-labview Feature #25024 (Resolved): Reference amplitude and phase
- We need to be able to plot the reference line magnitude and phase and to plot FFT as well.
- 04:20 AM llrf-SSR1-2Cavity-firmware Bug #25019 (New): Cavity signal on SSR1-8 has fast spikes.
- I tried closing the feedback loop but saw a very noisy drive signal. I then zoomed in on the cavity signal as saw fa...
- 02:38 AM llrf-hwr-2cavity-firmware Feature #24921 (New): Beam Loading Compensation and DAQ modes
- We need three modes of operation and DAQ. Pulsed mode and CW are spelled out in some TRS but DAQ for Beam Loading Co...
- 11:30 AM PIP II IT SSR1 Task #24741: Get LV installed on needed consoles
- I suggest that we have a single folder with all up to date LV executables. I will email Darren about a central locat...
- 11:26 AM llrf-ssr1-fe-software-acnet Bug #24894: When LV is turned off, code also stops running.
- Please update.
Also available in: Atom