- Registered on: 01/18/2019
- Last connection: 04/09/2021
- 02:17 PM Runs Task #25702: DR1_Run_10
- Measured transmission from from port 14 to port 3 (low frequency amp) and from port 14 to port 16 (high frequency amp...
- 01:29 PM Runs Task #25702: DR1_Run_10
- Cryo PC settings and Data:
BF software setup for data capturing. Data: Temperature and CryoSystem Logs, stored local...
- 12:46 PM Runs Task #25702: DR1_Run_10
- About Fluxgate Sensors and they references.
We have total of 5 FGs inside Dewar, they combined in three groups and r...
- 05:55 PM Runs Task #25702: DR1_Run_10
- 4/02/2021 Friday DR closed, pump down started;
- 05:07 PM Runs Task #25702: DR1_Run_10
TE3RI001 – Center, 3GHz, new Transmon, antennas 31.75mm both; transmon chip hanging from the top, centered...
- 12:23 PM Runs Bug #25198: DR2_Run_1
- EXT control of Still Heater was not working. Workflow:
Resistance of heater (with cable 3wiring, from wires discon...
- 07:20 PM Runs Bug #25198: DR2_Run_1
- Attempt fresh installation of BF software on new PC. This fix issue with TC400 communications.
Planning to cleanup d...
- 11:22 AM Runs Bug #25198 (New): DR2_Run_1
- Test run initiated after BF2 refrigerator was modified: we got it lowered down.
After first start of PC noticed issu...
Also available in: Atom