HWR 8 Channel Signal Conditioning Chassis
Test results and photos are attached
Test results were conducted by Philip and I in the Lab. Attached are the results.
- Estimated time set to 16.00 h
The 8 cavity signal conditioning chassis testing is complete. The pressure transducer
signal's dynamic range was corrected to match the ADC's full range.
The chassis is ready for installation at CMTF.
The corrected dynamic range modification and final test results shown in new results document.
Updated with resistor value for R4 = 1.21K. 7/27/20
- % Done changed from 90 to 100
- Status changed from Work in progress to Resolved
Philip installed on 7/31/20
Also available in: Atom