Milestone #17719: Booster operational test of one crate alongside existing BPM system
Verify DAQ operation with at least 4 modules
#1 Updated by John Diamond over 2 years ago
- % Done changed from 0 to 20
It appears the A32 Span register is not a number of bytes but some larger quantity (possibly 4-byte words). Reduced the span size that gets programmed into the modules by DAQSystem to 0x00400000. Also changed the first module's A32 base address to 0x0 and kept the step size to 0x01000000 to ensure there's plenty of address space between modules.