Upgrade of MFC system for 20 Hz
I have implemented a change to the front-end software for the MFC system to allow for 20 Hz operation in a stable fashion. The change was to reduce the number of points in the tables to 2120 from 5120. This will restrict the pulse width to 2.1 ms from 5 ms. The pulse width restriction occurs in the spreadsheet by limiting the flat-top time and the gate widths to 2.1 ms. The spreadsheet and code changes have been committed to cvs and the RFQ is now running production.