Configure R:TOR703 for g-2 operation
Configure S60TOR / R:TOR703 to operate with ABC03 firmware ADC
#1 Updated by John Diamond over 3 years ago
E-mail from Aisha:
With the new firmware, please have the front end set the following registers at startup. There will be no need for the averaging and current readouts as well as the dynamic gate width. (SYNC_OUT mode) 0x115 = x0 (GateWidth , 12us) 0x2BA = x5DC (#of RawSamples) 0x20C = x4000 (2^# = DecRate) 0x2B4 = x0 (#of AvgSamples) 0x2B7 = x4000 (SYNC delay) 0x2C0 = x0
#3 Updated by John Diamond over 3 years ago
Ning deployed the new firmware on Friday. Deployed my new startup script and latest VMEInt build to S60TOR on Sunday.
Implemented a command in VMEIntCommandsADC for setting the raw sample size.
Verified the list of registers are being set properly with one excepetion:I think Aisha got 0x2b4 and 0x2b7 reversed. For now I have them configured as:
- 0x2b4 = 0x4000
- 0x2b7 = 0x0
Will confirm with her on Monday.