Update Pxie Gate Width Calibration Mode
The gate width calibration mode requested by Engineering is updated to reflect the change of gate width from a 2B to a 4B register. Acnet devices for each ADC board are created.
#1 Updated by Roger Tokarek over 3 years ago
To enter calibration mode, in a pxint telnet session for example, the user types: gwcenable adcMapping, "microseconds"
gwcenable 0, "64.5"Microseconds is a decimal value and must be in quotes. The first parameter, adcMapping, indicates which ADCs are set to calibration mode. See the Acnet Devices Table below. Bug, adcMapping does not match ADC #1.
To disable calibration mode - that is to return to normal operation:
gwcdisable 0The user needs to remember to disable the calibration mode for the particular ADC they were working with.
Help is available:
See D150, pxie pages 2-9 for gate widths by ADC and page 10 for a one page list of all gate widths. (July 27, 2016)